Research on rapid prediction model of rice moisture content based on near infrared spectroscopy
Objective:In order to established a non-destructive, rapid and efficient method for detecting the moisture content of rice. Methods:This study, 161 rice samples were collected from 5 different regions were studied by near infrared spectroscopy combined with stoichiometry. By eliminating abnormal spectra and preprocessing the spectra, the prediction model of rice moisture content was established by partial least squares regression. Results:15 abnormal spectrum samples were eliminated using the method of principal component analysis combined with mahalanobis distance. The best spectral pretreatment was to eliminate the constant offset. The prediction model R2CAL established in the training set was 0.994 3, root mean square error of calibration (RMSEC) was 0.21%, R2CV was 0.993 6, and root mean square error of cross validation (RMSECV) was 0.32%, which indicated that the cross-validation of the prediction model had high accuracy in predicting sample moisture content. The prediction model was tested with the validation set samples. The validation determination coefficient R2VAL of the model validation set was 0.980 1, the root mean square error of prediction (RMSEP) was 0.36%, and the relative percent deviation (RPD) was 7.14, which indicated that the prediction model had high prediction accuracy for the unknown samples. The P-value (two-sided) of the mean equation T-test of the measured and predicted values of the samples in the validation set was 0.879, and the difference between the measured and predicted values of the samples in the validation set was not significant, indicating that the prediction results of the prediction model were highly reliable. The error between the predicted value and the measured value of the verification set samples was within ±1%, and more than 90% were within ±0.5%. Conclusion:The established rice moisture prediction model can be applied to actual production, and it provides a non-destructive, rapid and high-accuracy detection method for the inspection of rice harvesting and storage.
Du, LU; Jian-bo, TANG; Tai-ling, JIANG; Zhong-ai, CHEN; and Mu, PAN
"Research on rapid prediction model of rice moisture content based on near infrared spectroscopy,"
Food and Machinery: Vol. 38:
2, Article 9.
Available at: https://www.ifoodmm.cn/journal/vol38/iss2/9
 吴中华,王珊珊,董晓林,等.不同温度及含水率稻米籽粒加工过程破裂载荷分析[J].农业工程学报,2019,35(2):278-283.WU Zhong-hua,WANG Shan-shan,DONG Xiao-lin,et al.Analysis of rice compression fracture load in processes with various temperatures and moisture content[J].Transactions of the Chinese Society of Agricultural Engineering,2019,35(2):278-283.
 吴晓娟,吴伟,高崇明,等.压盖导流通风降温技术对储藏稻谷品质的影响[J].食品与机械,2016,32(6):132-135.WU Xiao-juan,WU Wei,GAO Chong-ming,et al.Effect of ventilation cooling technology by cover surface and guiding flow on paddy quality during storage[J].Food & Machinery,2016,32(6):132-135.
 YUAN L P.Development of hybrid rice to ensure food security[J].Rice Science,2014,21(1):1-2.
 中华人民共和国国家统计局.中国统计年鉴[M].北京:中国统计出版社,2020:12.The National Bureau of Statistics of the Peopleʼs Republic of China.China statistical yearbook[M].Beijing:China Statistics Press,2020:12.
 FAO.World food and agriculture-statistical year book 2020[M].Rome:[s.n.],2020:11-13.
 褚小立.近红外光谱分析技术实用手册[M].北京:机械工业出版社,2016.CHU Xiao-lin.The near infrared spectral analysis technology and practical manual[M].Beijing:Mechanical Industry Press,2016.
 CAPORASO N,WHITWORTH M B,FISK I D.Near-Infrared spectroscopy and hyperspectral imaging for non-destructive quality assessment of cereal grains[J].Applied Spectroscopy Reviews,2018,53(8):667-687.
 GUO Y,NI Y,KOKOT S.Evaluation of chemical components and properties of the jujube fruit using near infrared spectroscopy and chemometrics[J].Spectrochimica Acta Part A:Molecular and Biomolecular Spectroscopy,2016,153:79-86.
 SHAO W,LI Y,DIAO S,et al.Rapid classification of Chinese quince(Chaenomeles speciosa Nakai)fruit provenance by near-infrared spectroscopy and multivariate calibration[J].Analytical and Bioanalytical Chemistry,2017,409(1):115-120.
 RADY A,ADEDEJI A.Assessing different processed meats for adulterants using visible-near-infrared spectroscopy[J].Meat Science,2018,136:59-67.
 MAMANI LINARES L W,GALLO C,ALOMAR D.Identification of cattle,llama and horse meat by near infrared reflectance or transflectance spectroscopy[J].Meat Science,2012,90(2):378-385.
 鞠兴荣,后其军,袁建,等.基于近红外光谱技术测定稻谷含水量研究[J].中国粮油学报,2015,30(11):120-124.JU Xing-rong,HOU Qi-jun,WU Jian,et al.Determination of moisture content of rice based on the near infrared spectroscopy[J].Journal of Chinese Cereals and Oils Association,2015,30(11):120-124.
 杨学文.近红外光谱分析发快速测定稻谷常规化学指标[J].粮油食品科技,2016,24(5):65-69.YANG Xue-wen.Rapid determination of rice conventional chemical index by near infrared spectroscopy[J].Science and Technology of Cereals,Oils and Foods,2016,24(5):65-69.
 ZHANG F,LIU J,WANG Z.Analysis of oil yield from oil shale minerals based on near-infrared spectroscopy with least squares support vector machines[J].Chemical Journal of Chinese Universities,2016,37(10):1 792-1 798.
 ZHAN X,ZHU X,SHI X,et al.Determination of hesperidin in tangerine leaf by near-infrared spectroscopy with SPXY algorithm for sample subset partitioning and Monte Carlo cross validation[J].Spectroscopy and Spectral Analysis,2009,29(4):964-968.
 杜尚广.基于近红外光谱技术快速评价芸苔属种子活力[D].南昌:南昌大学,2015:38-39.DU Shang-guang.Rapid assessment of seed vigor of Brassica based on near infrared spectroscopy[D].Nanchang:Nanchang University,2015:38-39.
 CHEN L,YANG Z,HAN L.A review on the use of near-infrared spectroscopy for analyzing feed protein materials[J].Applied Spectroscopy Reviews,2013,48(7):509-522.
 WEYER L.Practical guide to interpretive near-infrared spectroscopy[M].[S.l.]:CRC Press,2007:57.